Bought used HAMEG HM605 oscilloscope with trigger problem. Tested it with sine wave and yes, cannot trigger it nor with automatic trigger neither manually. Decided to resolder all contacts near the trigger switch. Luckily it was easy accessible, just had to remove cover. After resolder next test revealed that sometimes trigger works, and sometimes not. Just a slight touch (no pressing) on external trigger button seems to be cause. So I resoldered all the contacts of ext trigger switch and some around and triggering started working as intended. Fixed.
Fig 4.2.2 Capacitor Voltage
Fig 4.2.3 Capacitor Current
Fig. 4.4.2 Current Through an Inductor